Takahashi, Sato, Horiuchi (Mirapro), Miyo
We confirmed that F3 was installed propery except the high sensor noise as mentioned in klog#15906.
We measured TF for F3 as we had done for BF in klog#15786.
The resonant frequency of F3 was 0.22 Hz as shown in the attached figure.
Measurement file is /users/Miyo/dropbox/LOG/210203/f3_tuning.xml.
Please note that we have not applied the new calibration factor for F3 filter as reported in klog#15891 for this TF plot. We used the previous one (15891)
Anyway, we will prepare to install the F2 filter.