Related work: klog19107
I checked the spectra of ITMX and ETMX payload (as for tower part, please see Ushiba-san's work).
In MN DAMP spectrum, there were three strange peaks around 0.7-1.1 Hz (surrouded by blue circle in Fig1).
*Solid line and Dashed line show the spectrum after/before finalization work.
With Ushibasan's advice, I checked the spectrum of BF and found that there were peaks at the same frequency (Fig2). So this may affects to peaks shown in Fig1.
Seeing the sensor spectrum, there are similarly peaks for H1, H2, H3 (Fig3). So we can say and are well balanced.
Besides this (ETMX IM and MN, ITMX IM), there was no problem.